> Fundamentals of Fluid Mechanics > Steady Incompressible Flow > Flow around Bodies > Steady Flow of Compressible Fluids > Examples of Transient Flow > Hydraulic Fluid Energy Machines > Components in Piping Systems and Plant Design > Fluidic Experimental Plants > Hydraulics for Civil Engineering
 Products  |  Fluid Mechanics  |  Steady Incompressible Flow  |  HM 500  |  HM 500.13
 HM 500.13 Orifice Plate Flow Meter with Transducer
 Technical Description

The orifice plate flow meter is installed in the water circuit of the HM 500 trainer. The flow rate measurement is based on the differential pressure method.
The orifice plate narrows the cross-section in the tube. The constriction of the cross-section causes an increase in velocity which results in a measurable differential pressure. A transducer based on variable capacitance (diaphragm method) is used to measure the differential pressure. It is indicated on a display.
Taking the orifice geometry into account, the flow rate can be calculated from the decrease in pressure using Bernoulli’s principle and the Continuity law.

Learning Objectives / Experiments

- familiarisation with the operating principle
* continuity law and Bernoulli's principle
* transducer with capacitive sensing element
- flow rate measurement
- plotting a pressure loss curve
- comparison with other flow meters

 Features

* Orifice Plate Flow Meter with electronic differential

pressure transducer for flow rate measurement1

 Specification [1] orifice Plate Flow Meter with electronic differential pressure transducer for flow rate measurement as accessory for trainer HM 500[2] operation based on the differential pressure method[3] display indicating differential pressure[4] connections to facilitate pressure loss measurement with the HM 500[5] connections to supply auxiliary power via the HM 500[6] vertical and horizontal installation possible Technical Data Orifice plate- material: brass- diameter: 18,5mm, with 45° chamferTransducer measuring range: 0...500mbarAuxiliary power: 24VDCPipe connections: DN 32 Dimensions and Weight LxWxH: 820x200x250mmWeight: approx. 8kg Scope of Delivery 1 orifice plate flow meter with transducer1 set of instructional material Order Details 070.50013  HM 500.13  Orifice Plate Flow Meter                      with Transducer